Water Damage to Cardiovascular Ultrasound Machine
Envista was called in to investigate fugitive water that impacted a Cardiovascular Ultrasound (echo) machine located within a room that had a punctured water pipe.
The insured was concerned that the water and moisture may have caused accelerated corrosion and future contamination-related reliability issues to the echo machine.
Envista’s equipment loss consulting team carried out an internal inspection by visually examining internal components, such as the PCBs (printed circuit boards), continuity testing, and wipe sample analysis.
The inspection did not reveal presence of water, physical damages, or flash oxidation on the surface of circuit-level components and modules. Other than the presence of watermarks on the exterior surfaces of some components observed during Envista’s inspection, no damages were confirmed.
The cost to replace part of the equipment and decontaminate other affected parts would come to a total of $11,000, far less than the claimed $200,000 for replacement of the entire equipment.
Nuestros consultores están listos para ayudar.